Department of Anesthesiology, Intensive Care and Pain Therapy, Onze Lieve Vrouwziekenhuis, Aalst, Belgium

Department of Anesthesia, Stanford School of Medicine, Stanford, California, USA

Department of Anesthesiology, Sint-Jan Hospital, Brussels, Belgium

Department of Anesthesiology, Intensive Care and Pain Therapy, Catharina Hospital, Eindhoven, The Netherlands

Department of Clinical Pharmacy, Catharina Hospital, Eindhoven, The Netherlands

Department of Anesthesiology, University of Ghent, Ghent, Belgium

Department of Anesthesiology, Northwestern University Medical School, Chicago, Illinois, USA

Abstract

Background

The wide range of fresh gas flow - vaporizer setting (FGF - F_{D}) combinations used by different anesthesiologists during the wash-in period of inhaled anesthetics indicates that the selection of FGF and F_{D }is based on habit and personal experience. An empirical model could rationalize FGF - F_{D }selection during wash-in.

Methods

During model derivation, 50 ASA PS I-II patients received desflurane in O_{2 }with an ADU^{® }anesthesia machine with a random combination of a fixed FGF - F_{D }setting. The resulting course of the end-expired desflurane concentration (F_{A}) was modeled with Excel Solver, with patient age, height, and weight as covariates; NONMEM was used to check for parsimony. The resulting equation was solved for F_{D}, and prospectively tested by having the formula calculate F_{D }to be used by the anesthesiologist after randomly selecting a FGF, a target F_{A }(F_{At}), and a specified time interval (1 - 5 min) after turning on the vaporizer after which F_{At }had to be reached. The following targets were tested: desflurane F_{At }3.5% after 3.5 min (n = 40), 5% after 5 min (n = 37), and 6% after 4.5 min (n = 37).

Results

Solving the equation derived during model development for F_{D }yields F_{D}=-(e^{(-FGF*-0.23+FGF*0.24)}*(e^{(FGF*-0.23)}*F_{At}*Ht*0.1-e^{(FGF*-0.23)}*FGF*2.55+40.46-e^{(FGF*-0.23)}*40.46+e^{(FGF*-0.23+Time/-4.08)}*40.46-e^{(Time/-4.08)}*40.46))/((-1+e^{(FGF*0.24)})*(-1+e^{(Time/-4.08)})*39.29). Only height (Ht) could be withheld as a significant covariate. Median performance error and median absolute performance error were -2.9 and 7.0% in the 3.5% after 3.5 min group, -3.4 and 11.4% in the 5% after 5 min group, and -16.2 and 16.2% in the 6% after 4.5 min groups, respectively.

Conclusions

An empirical model can be used to predict the FGF - F_{D }combinations that attain a target end-expired anesthetic agent concentration with clinically acceptable accuracy within the first 5 min of the start of administration. The sequences are easily calculated in an Excel file and simple to use (one fixed FGF - F_{D }setting), and will minimize agent consumption and reduce pollution by allowing to determine the lowest possible FGF that can be used. Different anesthesia machines will likely have different equations for different agents.

Background

What fresh gas flow - vaporizer setting (FGF - F_{D}) combination should be used for a particular patient when starting the administration of potent inhaled anesthetics to reach a target end-expired concentration (F_{A}) after a predetermined time interval without excessively wasting potent inhaled anesthetic? The wide range of FGF - F_{D }combinations used by different anesthesiologists during the wash-in period of potent inhaled anesthetics indicates that the selection of FGF and F_{D }is based on habit and personal experience. Some anesthesiologists use a high FGF (to shorten the wash-in time constant of the anesthesia circle breathing system, and to avoid rebreathing that results in dilution of F_{D}), while others prefer to use a lower FGF in combination with a higher F_{D }(to compensate for the longer wash-in time constant, and to reduce agent consumption). While all anesthesiologists swiftly attain the target F_{A }(F_{At}) because FGF and F_{D }can be adjusted according to the measured F_{A}, it is unlikely that the particular FGF - F_{D }combination used was that with the least number of F_{D }and FGF adjustments and minimum waste. The use of high FGF, even for a seemingly brief period (5 min), may increase agent consumption above that of an ensuing one hour maintenance phase with a 1 L. min^{-1 }FGF

Instead of relying on personal preference, we hypothesize that very specific FGF - F_{D }combinations can be used in the individual patient to attain a F_{At }within a specified time interval by using an empirical model of the kinetics of inhaled anesthetics during wash-in. Simple, easy to remember FGF - F_{D }combinations construed from these models could reduce agent consumption while not distracting the anesthesiologist from other tasks during the induction period of anesthesia

While kinetics of inhaled anesthetics in the anesthesia circle system have already been modeled using mass balances _{2 }with an ADU - AS/5^{® }anesthesia machine (Anesthesia Delivery Unit, General Electric, Helsinki, Finland) during the first 5 min of the anesthetic, and prospectively tested whether it allows the anesthesiologist to select a FGF - F_{D }combination to attain a F_{At }of the inhaled anesthetic within a specified time interval (1 - 5 min) after turning on the vaporizer.

Methods

Part I. Model development

After obtaining IRB approval (OLV Hospital, Aalst, Belgium) and written informed consent, 50 ASA physical status I or II patients presenting for plastic, urologic, or gynecologic surgery were enrolled. All patients received oral alprazolam (0.5 or 1.0 mg) 1 h before the scheduled start of surgery. After preoxygenation (8 L. min^{-1 }O_{2 }FGF for 3 min), propofol (3 mg. kg^{-1}), rocuronium (0.7 mg. kg^{-1}), and sufentanil (0.1 μg. kg^{-1}) were administered intravenously. After tracheal intubation, ventilation was mechanically controlled by an ADU anesthesia machine. Tidal volume and respiratory rate were set fixed at 500 mL and 10 breaths. min^{-1}, respectively.

In a particular patient, one (fixed) O_{2 }FGF and desflurane F_{D }combination was used, with the FGF ranging from 0.5 tot 5 L. min^{-1 }and F_{D }from 6 to 18%; this FGF - F_{D }combination was chosen randomly (random function in excel). Preliminary trials indicated what combinations were likely to lead to a desflurane F_{A }less than 1% after 2 min or less than 2% after 5 min, and these were not considered.

Inspired and expired gases were analyzed by a multigas analyzer (Datex-Ohmeda Compact Airway Module M-CAiOV^{®}, Datex-Ohmeda, Helsinki, Finland) and downloaded into a spreadsheet every 10 seconds. Gases were sampled at the distal end of the endotracheal tube using a piece of sampling tubing placed through an Arndt Multi-Port Airway Adapter^{® }(Cook Medical Inc., Bloomington, IN). Gases sampled by the gas analyzer were redirected to the anesthesia circuit via the expiratory limb. The study was terminated after 5 minutes, or earlier when the end-expired desflurane concentration had reached 8%. All values above 8% were eliminated from further analysis. The 5 min period was somewhat arbitrarily defined as the wash-in period because it encompasses (1) anesthesia circuit wash-in; (2) FRC wash-in; (3) early uptake by the VRG; (4) and the waning effects of propofol after about 5 min.

All measurements were done with the same anesthesia machine and gas analyzer. The ADU^{® }circle system volume is 3.4 L. The fresh gas flow inlet is located distal to the inspiratory valve. Fresh gas flow compensation is used to compensate for the inspiratory fresh gas flow during inspiration. The vaporizer output was measured at the common gas outlet by the same gas analyzer and compared with the dial setting using linear and non-linear regression because (1) the vaporizer output may not match the dial setting; (2) we wanted to be able to generalize the results to other ADU^{® }units; and (3) we wanted to exclude that certain performance error patterns could be related to systematic vaporizer error. Based on 229 measurements in 48 patients in this study, the actual desflurane vaporizer output (%) could be described as -0.72 + 1.075*dial setting (r^{2 }= 0.98); the vaporizer's output tended to increase with lower FGF (Figure

Vaporizer performance error

**Vaporizer performance error**. Vaporizer performance error (PE) with different fresh gas flows (FGF). PE (%) = 100*(measured output - vaporizer dial)/vaporizer dialed. Thick line = linear regression, with PE = 4.101-1.902*FGF (r^{2 }= 0.23); grey line = non-linear regression (third order polynomial), with PE = 6.990 - 5.959*FGF + 1.099*FGF^{2 }- 0.05192*FGF^{3}.

A model was build to relate F_{A }to FGF, F_{D}, time, and the following patient covariates: age, height, and weight. A constant ventilation allowed us to at least standardize circuit and FRC wash-in; after 5 min, ventilation can easily be adjusted to the desired end-expired CO_{2 }concentration. All values before 1 min were deleted because zero values are hard to work with mathematically, and because the model was only supposed to model the FGF - F_{D }- time relationship between 1 and 5 min. Initial model building and parameter exploration were done in Excel using Solver^{® }(Microsoft, Seattle, WA). The initial choice of mathematical functions was guided by three assumptions. First, because F_{A }rises exponentially, a one exponential function was used to describe the rise of F_{A}. Second, the effect of a higher F_{D }was modeled as curvilinear (assuming, for example, that doubling F_{D }would lead to a doubling of F_{A}). Third, the effect of FGF was modeled with a single exponential (increasing as FGF is lowered). By trial and error, these and additional functions were added, deleted, modified, etc. to minimize the sum of least squares (difference between measured and predicted F_{A}) using Solver (Excel). Residuals were plotted against height, weight, and age to search for covariate effects by visual inspection and by linear regression. Finally, the model was tested for parsimony using NONMEM's Minimum Objective Function (ICON Development Solutions, Dublin, Ireland).

Part II. Prospective testing

The model equation derived in part I was solved for F_{D }using Mathematica (Mathematica for Windows, Version 4.0, Wolfram Research Inc, Champaign, IL). The resulting equation predicts the FGF - F_{D }combinations the anesthesiologist can use to attain a F_{At }within the same time interval used during model development (i.e., between 1 and 5 min) using a single F_{D }and FGF setting, and takes into account the covariate effects derived during model building (see results section for actual equation).

Management of the patients during prospective testing only differed in the manner in which FGF and F_{D }were selected. Patients received desflurane in O_{2 }with the goal to reach a F_{At }of 3.5% after 3.5 min (n = 40), 5% after 5 min (n = 37), or 6% after 4.5 min (n = 37). The number of patients was chosen based on prior experience. After entering the time and F_{At }as well as significant patient covariates in the equation, the equation describes all possible FGF - F_{D }combinations that reach the F_{At }at the desired time for a patient with the particular characteristics entered into the equation. The fixed FGF that was going to be used in the individual patient was randomly selected (using Excel's random function) and entered in the equation, yielding the F_{D }to be used. Because the resolution of the desflurane vaporizer is 0.5%, the nearest value was chosen. FGF values requiring an F_{D }above the vaporizer limit (18%) obviously could not be tested.

To allow us to compare model performance between the three subgroups (3.5% after 3.5 min, 5% after 5 min, or 6% after 4.5 min), the performance error (PE) for each patient was calculated as 100*((F_{A }measured - F_{A }predicted)/F_{A }predicted), and the absolute performance error (APE) as the absolute value of PE. Next, for each subgroup, the following were examined: (1) bias and accuracy, using the median performance error (MDPE, median of all PE) and median absolute performance error (MDAPE, median of all APE)

Results

Part I. Model derivation

Patient demographics are presented in Table _{A }best:

Patient demographics, presented as mean (standard deviation)

**Group**

**Number**

**Age (years)**

**Weight (kg)**

**Height (cm)**

Model derivation group

50

52 (15)

70 (14)

166 (9)

3.5% after 3.5 min group

40

46 (16)

72 (12)

170 (9)

5% after 5 min group

37

53 (18)

77 (16)

170 (8)

6% after 4.5 min group

37

52 (16)

73 (14)

169 (9)

with Ht = height (cm), Time = time after turning on the vaporizer (min), and with F_{A}, F_{D }and FGF expressed in %, %, and L. min^{-1}, respectively. The measured versus model predicted F_{A }are presented in Figure ^{(-FGF*-0.23)}+2.49*(1-e^{(-FGF*0.24)})*0.39*F_{D}), will cause the F_{A }to decrease again at high flows, an effect caused by a decreased vaporizer output with high FGF due to cooling.

Model derivation: measured versus model predicted end-expired desflurane concentrations (F_{A}, %)

**Model derivation: measured versus model predicted end-expired desflurane concentrations (F**_{A}**, %)**. Thick line = identity line; lines above and below are + or - 20% deviation from identity line.

Model derivation: final plot of residual error (modeled minus measured F_{A}) versus patient covariates

**Model derivation: final plot of residual error (modeled minus measured F**_{A}**) versus patient covariates**. A = age, B = height, and C = weight. There was no (linear) correlation with age (r^{2 }= 0.00), height (r^{2 }= 0.01 after implementing it as a covariate in the model) and weight (r^{2 }= 0.00).

Part II. Prospective testing

Solving the above derived equation for F_{D }yields the following equation:

where Time = desired time lag after turning on the vaporizer to reach F_{At }(any value between 1 and 5 min); and Ht = patient height in cm. After entering F_{At}, Time, Ht, and chosen FGF, the corresponding F_{D }to reach F_{At }within the desired time lag can be calculated. Because any FGF can be chosen, this will result in a virtually infinite number of FGF - F_{D }combinations. Patient demographics for each subgroup are presented in Table _{A }and age, but not with height (effect incorporate in model) and weight (Figure

Model performance

**Model performance**. Performance Error (PE, %, closed circles, group 3.5% after 3.5 min = A; group 5% after 5 min = C, and group 6% after 4 min = E) and Absolute Performance Error (APE, %, open circles, group 3.5% after 3.5 min = B, group 5% after 5 min = D, and group 6% after 4 min = F), with their respective linear regression line (hatched), polynomial fit (grey line), and median performance error (MDPE, continuous black line, A, C, and E) and median absolute performance error (MDAPE, continuous black line, B, D, and F).

Prospective testing

**Prospective testing**. Linear regression between covariates (age, height, and weight) and residual error (predicted minus measured end expired desflurane concentration) in the 3.5% after 3.5 min group (A, B, and C), the 5% after 5 min group (D, E, and F), and the 6% after 4.5 min group (G, H, and I).

Discussion

An empirical equation can be derived that predicts FGF - F_{D }combinations that attain a F_{At }of an inhaled anesthetic after a predefined time lag. With modern computing power such an equation can easily be entered into an Excel file to calculate the required F_{D }for any chosen FGF. While our current model will need to be refined and tested for different patient populations, agents, carrier gases, and anesthesia machines, our results prove the concept.

The concept of using an equation to predict the required F_{D }for any chosen FGF to help the anesthesiologist attain and maintain a F_{At }is not new. Three decades ago, Lowe described a "general anesthetic equation" that describes the FGF - F_{D }relationship in a circle breathing system _{D }relationship during the maintenance phase.

The MDPE (-2.9, -3.4, and -16.2%) as well as the MAPDE (7.0, 11.4, and 16.2%) are within the limits deemed acceptable for target controlled infusion systems for intravenous anesthetic agents (MDPE <10-20% and MDAPE 20 - 40%) _{A }with lower FGF because increased rebreathing causes the effect of the (unpredictable) amount of uptake on the composition of the inspired mixture to become more pronounced. During modeling, attempts are made to improve the degree of misspecification by taking the effect of covariates into account. However, only height significantly decreased the minimum objective function. Weight (within the range encountered in the study population) has previously been shown

- not

Other limitations exist. The wash-in model may not be applicable when spontaneous ventilation is allowed immediately following intravenous induction of anesthesia, because the irregular and inconsistent breathing at that time does not allow the acquisition of reliable end-expired concentrations. This also is an issue for modern anesthesia machines that use automated closed-loop end-tidal feedback administration of inhaled agents. Also, the model is limited to a maximum end-expired concentration of 8% - higher concentrations might be needed in some patients, but this comes at a risk of irritating the airway.

An empirical equation that describes the FGF - F_{D }relationship may have some interesting applications. It can be used to predict and depict the course of F_{A }in the individual patient over a wide range of FGF - F_{D }settings during the first 5 min of an anesthetic. In figure _{A }resulting from different FGF - F_{D }combinations after 1, 2, 3, 4, and 5 min (multi-colored graphs, bottom to top, respectively; more detail on how this graph was derived can be found in the Appendix). The equation can further be used to calculate the FGF - F_{D }combinations that attain a F_{At }after a predefined time interval. If the F_{At }would be 4.5% (the light blue surface), the required FGF - F_{D }combinations that attain 4.5% after 1, 2, 3, 4, and 5 min can be found by calculating the intersection between the multicolored surfaces and the light blue surface. These intersections are presented in Figure _{At }(i.e., it illustrates the FGF below which vaporizer output becomes inadequate to attain the F_{At }within the specified time interval).

Visual presentation of model-based desflurane F_{A }course in a fictive 176 cm tall patient

**Visual presentation of model-based desflurane F**_{A }**course in a fictive 176 cm tall patient**. The multi-colored graphs (A) describe the predicted F_{A }resulting from different FGF - F_{D }combinations after 1, 2, 3, 4, and 5 min (bottom to top, respectively). The intersection of these surfaces with the F_{At }surface, in this example 4.5% desflurane (light blue horizontal surface), yields the corresponding FGF - F_{D }combinations that attain that F_{At }(B). The asterixes (*) indicate the lowest possible FGF that can be used to attain 4.5% desflurane at 1, 2, 3, 4, and 5 min because of the maximum 18% vaporizer setting. More details are provided in the Appendix.

Conclusion

An empirical model is described that allows the derivation of a formula for each type of anesthesia machine that predicts the FGF - F_{D }combinations in a circle breathing system that attain a target end-expired agent concentration in a particular patient within a specified time interval (1 - 5 min) after turning on the vaporizer. The sequences are easily calculated in an Excel file and simple to use (one fixed FGF - F_{D }setting), and have the potential to minimize agent consumption and reduce pollution by allowing to determine the lowest possible FGF that can be used.

Appendix: Derivation of Figure

According to the model, F_{A }= (1/(Ht*0.1))*(2.55*FGF+1348*(1-e^{(-FGF*-0.23)}+2.49*(1-e^{(-FGF*0.24)})*0.39*F_{D})*(0.03*(1-e^{(Time/-4.08)}))) with Ht = height (cm), Time = time after turning on the vaporizer (min), and with F_{A}, F_{D }and FGF expressed in %, %, and L. min^{-1}, respectively.

Let us examine how the desflurane F_{A }would evolve for a 176 cm tall patient over the entire FGF-F_{D }range studied during model development. The surface describing desflurane F_{A }after 3 min is calculated by entering the following parameters into the above formula: Ht = 176; Time = 3; and a sufficient number of FGF - F_{D }combinations to allow the reconstruction of a surface in Sigmaplot (Systat Software Inc, San Jose, CA, USA). The multicolored surface in Figure _{A }values after 3 min.

Multicolored surface describing desflurane _{A }after 3 min for a 176 cm tall patient over the FGF - F_{D }range studied

**Multicolored surface describing desflurane F**_{A }**after 3 min for a 176 cm tall patient over the FGF - F**_{D }**range studied**. The FGF - F_{D }combinations that result in a F_{At }of 4.5% (light blue horizontal surface) after 3 min are found by calculating the intersection between this target (the light blue surface) and the multicolored surface; these FGF - F_{D }combinations are presented in B. The asterix (*) indicates the lowest possible FGF that can be used to attain 4.5% desflurane at 3 min because of the maximum 18% vaporizer setting.

To determine which FGF - F_{D }combinations result in a F_{At }of e.g. 4.5% after 3 min, the intersection has to be calculated between this target (the horizontal light blue surface) and the multicolored surface that describes the F_{A }values with the entire FGF-F_{D }range after 3 min (Figure _{At }= 4.5 in the formula F_{D}=-(e^{(-FGF*-0.23+FGF*0.24)}*(e^{(FGF*-0.23)}*F_{At}*Ht*0.1-e^{(FGF*-0.23)}*FGF*2.55+40.46-e^{(FGF*-0.23)}*40.46+e^{(FGF*-0.23+Time/-4.08)}*40.46-e^{(Time/-4.08)}*40.46))/((-1+e^{(FGF*0.24)})*(-1+e^{(Time/-4.08)})*39.29). This intersection line is shown in Figure _{D }combinations that result in an F_{A }of 4.5% after 3 min in a 176 cm tall patient. It also illustrates that it is not possible to achieve this target with a FGF lower than 0.7 L. min^{-1 }because F_{D }would have to be higher than the maximum 18%. Note that the curve increases at high FGF because vaporizer output decreases due to excessive cooling.

Multicolored surface describing desflurane F_{A }after 5 min for a 176 cm tall patient over the FGF - F_{D }range studied

**Multicolored surface describing desflurane F**_{A }**after 5 min for a 176 cm tall patient over the FGF - F**_{D }**range studied**. The FGF - F_{D }combinations that result in a F_{At }of 4.5% (light blue surface) after 5 min are found by calculating the intersection between this target (the light blue surface) and the multicolored surface; these FGF - F_{D }combinations are presented in B. The asterix (*) indicates the lowest possible FGF that can be used to attain 4.5% desflurane at 5 min because of the maximum 18% vaporizer setting.

Similar graphs can be developed for different time intervals (between 1 and 5 min). The multicolored surface in Figure _{A }after using a range of FGF - F_{D }combinations for 5 min; the intersection between this surface and the target 4.5% surface (horizontal light blue surface) thus describes the virtually infinite number of FGF - F_{D }combinations that result in an F_{A }of 4.5% after 5 min in a 176 cm tall patient (Figure _{At }of 4.5%.

Competing interests

Jan Hendrickx has received speaker fees and reimbursement of travel expenses related to presentation of his work from GE, Abbott, and Baxter.

Authors' contributions

JH, HL and ADW conceptualized the idea, designed the study, and performed mathematical analysis. All authors participated in data analysis and interpretation and drafting of the manuscript. All authors read and approved the final manuscript.

Pre-publication history

The pre-publication history for this paper can be accessed here: